The signal of each EEG channel is analyzed during the recording and labeled accordingly (good or bad quality). Then the proportion of good quality EEG signals is computed and given through this metric.
How to read them
With a Dreem3 headband, you will have access to three values. These values pertain to sensors: Left Occipital sensor, Right Occipital sensor, and Frontal sensors respectively. With a Dreem3 headband, there is two values displayed for the frontal sensors.
How it works
Two seconds signal portions of raw data over 35-40 nights have been labeled by our team of sleep specialists (zero for bad quality, one for good quality). We trained our deep learning algorithm to automatically score the signal for each new night. The number displayed for each channel is the cumulative length of portions (i.e. a 2-second raw signal) scored as “of good quality” by the algorithm over the total length of a recording. Demander à Eden les guidelines du labeling.
A value below 70 is often related to a poor placing of the headband on the head. We advise you to review with your subject how to wear the headband and to switch to small adjustment strips.